I can add some to Kevin's reply. It is a Fusible resistor being used as
a sort of fuse. For safety you should use an exact replacement. If not,
then at least find a METAL FILM replacement. It is 1Kohm, 2Watt, 2%. If
yours is a plain 77 (not series II or III) then my sevice manual gives
the Fluke part number as 854687. It is part number FA8466 from
a manufacturer with FSC=23237, but I do not know which one this is.
> Hi, working on a Fluke 77 multimeter, on the volt/ohm input I have a
> blown resistor/capacitor/choke. Looks about the size of a 1 watt
> resistor with only two bands. One band is green the other is blue, can
> anybody identify it for me?
Professor Le Baron O. Ferguson Office: Sproul 2219
Department of Mathematics (909) 787-5003
Riverside CA 92521 FAX : (909) 787-7314